In this short episode Shahriar demonstrates an overview of the new Tektronix MSO58 8-Channel 6.25GS/s 2GHz Mixed-Signal Oscilloscope with up to 64-Bits of digital channels. The scope features independent ADC, data acquisition and memory per channel as well as FlexChannel architecture which allows each channel to act as either a 2GHz analog channel or an 8-Bit digital channel. The scope also features a brand new GUI interface on an HD 15.6″ display with significant performance enhancement and touch interface optimizations. A full review of the instrument will be provided in the future.
Archive for Equipment Review
In this episode Shahriar reviews the Tektronix RSA607A real-time spectrum analyzer which operates from 9kHz to 7.5GHz. The instrument has a 40MHz wide analysis window with a DANL of -160dBm. The built-in tracking generator is capable of measuring return loss, cable loss and DTF measurements without a need for an external coupler. This review is organized as follows:
01:03 – Model comparison and overview.
03:25 – Front & rear panel overview.
05:31 – Complete unit teardown & examination.
16:33 – Reverse engineering and experiments with a drone remote control transmitter.
39:27 – Characterizing and experimenting with an ISM band diplexer.
47:29 – Return loss measurements of a tunable band-pass filter.
53:23 – Concluding remarks.
In this episode Shahriar reviews the Deepace KC901V 6.8GHz handheld network analyzer. This battery-powered instrument is an RF multi-instrument integrating VNA, spectrum analyzer, field strength meter, and a low-frequency signal generator. It can also perform signal port vector measurement and 2-ports simple vector network analyzing (S11, S21). The review is organized as follows:
01:24 – Instrument overview & block diagram.
05:36 – Full teardown along with RF and digital board analysis.
14:30 – Power on, calibration and S11 measurement of an antenna.
23:46 – Calibration for S21 and measurement of a tunable filter plus phase shifter.
32:23 – Distance to fault measurement of an SMA cable chain.
36:24 – Spectrum analyzer measurements and limitations.
43:16 – Signal generator measurements and issues.
48:20 – Other features and concluding remarks.
50:07 – Giveaway!
In this episode Shahriar reviews the Rohde & Schwarz 4.0GHz Spectrum Rider FPH. The new design of the FPH series is aimed for ease of use in outdoor situations as well as offering several software features for data collection, analysis and debugging of standard wireless communication protocols. The review is organized as follows:
1:10 – Model comparison of R&S portable spectrum analyzers.
5:44 – Instrument hardware overview and GUI characteristics.
22:24 – Using the FPH Spectrum Rider to analyze unknown wireless signals including a multi-tone, QPSK modulated signal, AM/FM demodulation analysis and frequency hopping.
46:46 – FPH teardown and analysis.
55:33 – Overview of the Instrument View remote connection software.
57:42 – Concluding remarks.
In this episode Shahriar reviews the long awaited Keysight MXA Signal Analyzer (N9020B). The new X-Series Spectrum Analyzers from Keysight offer an entirely re-designed GUI interface which supports multiple tabs as well as multi-touch interaction. The review is organized as follows:
1:07 – Model comparison and overview of x-series spectrum analyzers.
7:08 – Instrument physical overview.
11:49 – New GUI overview and demonstrations.
22:44 – Instrument block diagrams and analysis.
28:46 – Brief look inside the MXA spectrum analyzer.
32:53 – Back-scattering communication experiment block diagram and description.
37:16 – Phase noise characterization of doubler and amplifier.
51:54 – Complete back-scattering experiment measurements and analysis.
1:04:45 – Analysis of broadband modulated signals by using the S-Series scope and MXA Signal analyzer at the same time.
1:13:56 – Concluding remarks.