In this episode Shahriar reviews the Deepace KC901V 6.8GHz handheld network analyzer. This battery-powered instrument is an RF multi-instrument integrating VNA, spectrum analyzer, field strength meter, and a low-frequency signal generator. It can also perform signal port vector measurement and 2-ports simple vector network analyzing (S11, S21). The review is organized as follows:
01:24 – Instrument overview & block diagram.
05:36 – Full teardown along with RF and digital board analysis.
14:30 – Power on, calibration and S11 measurement of an antenna.
23:46 – Calibration for S21 and measurement of a tunable filter plus phase shifter.
32:23 – Distance to fault measurement of an SMA cable chain.
36:24 – Spectrum analyzer measurements and limitations.
43:16 – Signal generator measurements and issues.
48:20 – Other features and concluding remarks.
50:07 – Giveaway!
In this short episode Shahriar takes a close look at an HP 83475B Lightwave Communication Analyzer / Oscilloscope which offers a 500MHz bandwidth on both the electrical and optical inputs. The sub-sampling scope has a minor issue where the CRT brightness knob has been disconnected internally.
The teardown of the scope reveals the sub-sampling ADCs, O/E front-end as well as CRT driver circuitry. The operation of the scope is demonstrated using various electrical and optical NRZ and multi-level signals.
In this episode Shahriar reviews the Rohde & Schwarz 4.0GHz Spectrum Rider FPH. The new design of the FPH series is aimed for ease of use in outdoor situations as well as offering several software features for data collection, analysis and debugging of standard wireless communication protocols. The review is organized as follows:
1:10 – Model comparison of R&S portable spectrum analyzers.
5:44 – Instrument hardware overview and GUI characteristics.
22:24 – Using the FPH Spectrum Rider to analyze unknown wireless signals including a multi-tone, QPSK modulated signal, AM/FM demodulation analysis and frequency hopping.
46:46 – FPH teardown and analysis.
55:33 – Overview of the Instrument View remote connection software.
57:42 – Concluding remarks.
In this short episode Shahriar shows a few items to give away! Please leave a comment and a person at random will be chosen. There will also be a Q/A session sometime in the future, please also leave your questions in the comment section.
The winners of the draw will be notified at a later date.
In this short episode Shahriar investigates the failure cause of a Keithley 2400 SMU. The unit powers on and responds to controls however produces no output signal. Further examination reveals that the VFD display shows no activity when the output is enabled; normally the measured output voltage and current are displayed. This failure points to a digital problem, potentially a failure in the analog board FPGA.
Upon close inspection the FPGA clocking circuitry appears to have a failed solder-joint. A quick re-flow of the affected part corrects the problem and the unit reports the appropriate readings. The unit is then upgraded to the latest firmware and calibrated using a Keithley 7510 DMM.
The Signal Path (TSP) is an electrical engineering video blog for industry professionals, students and hobbyists. TSP is a non-for-profit website dedicated to provide free education spanning a wide range of electrical engineering topics. Equipment reviews, tutorials and repair videos are posted regularly.