In this episode Shahriar reviews the long-awaited Keysight MXR-Series 8-Channel 6-GHz 16-GS/s Real-Time Oscilloscope. The Infiniium MXR-Series is the first oscilloscope to offer 6-GHz bandwidth and 16-GSa/s sample rate on every single one of its eight channels. It also offers excellent performance (ENOB up to 9.0, noise as low as 43 µVrms, dedicated ASICs) which exceeds that of the previous S-Series scopes.
This review is organized as follows:
00:00 – Introductions
01:02 – Instrument overview, front-panel and rear interfaces
4:09 – Full acquisition board teardown and analysis including analog front-end
19:29 – First power-on and impressions
20:48 – Comparison and experiments with the Digital Sampling FFT, DDC and RTSA modes
34:04 – Time-base accuracy, frequency counter and DVM experiments
38:30 – Real-Time Eye, Jitter analysis and Equalization capabilities 50:00 – Concluding remarks
In this episode Shahriar examines a failed Agilent 1169A 12GHz Smart Active Probe. The probe does not get recognized by the oscilloscope and therefore cannot be calibrated. The instrument uses an I2C interface to the probe to download the probe’s specifications and characteristics.
The probe teardown reveals a simple I2C EEPROM IC. After removing the IC it becomes clear that the chip is damaged and must be replaced. A duplicate probe is used to copy the EEPROM data onto a new IC which revives the probe. With the help of the community, the CRC checksum code is discovered and the probe is assigned a new serial number.
In this episode Shahriar presents an overview of a small portion of the vendors present on the exhibition floor at the International Microwave Symposium 2019. The IEEE MTT International Microwave Symposium (IMS) is the premier annual international meeting for technologists involved in all aspects of microwave theory and practice. It consists of a full week of events, including technical paper presentations, workshops, and tutorials, as well as numerous social events and networking opportunities. The symposium also hosts a large commercial exhibition. Co-located with IMS are the IEEE RFIC, IMBioC and ARFTG conferences.
In this episode Shahriar repairs a non-functional N4901B 13.5Gb/s BERT mainframe. This instrument is equipped with both the generator and analyzer modules which makes it fairly valuable. The instrument does not power on and does not react to the soft power button.
Teardown of the unit reveals a TD-Lambda Vega power supplies series which is a fully configurable switching power supply design. Even though the power supply in the instrument is completely custom, it is made from generic sub-blocks readily available on eBay. A replacement front-end is purchased to repair the power supply after detecting the problem.
The instrument also shows a minor problem with falsely detecting an overload condition on the delay control port which is by passed by installing an overwrite switch. The instrument performance is verified by generating and measuring various eye diagrams through a back-plane PCB board.
In this episode Shahriar reviews three mailbag items. There is also a new giveaway! Follow the instruction at the beginning of the video to enter. The reviews are organized at follows:
00:09 – Introduction & Giveaway Details
02:21 – Leo Bodnar Electronic – Fast Pulse Generator
These fast pulse generators are rated to below 25ps! The modules are examined and various microwave and design considerations are explored. The block diagram and functionality of the final output driver is also described in details. The units are used to measure the rise and fall time of various oscilloscopes ranging from 300MHz to 13GHz of analog bandwidth. You can purchase your own fast pulse generator here.
21:11 – SV1AFN GPS Disciplined Oscillator
This GPS disciplined oscillator not only provides frequency synchronization to the GPS clock, it also provides programmable outputs on two independent channels using a Silicon Labs synthesizer IC. The complete module acts as standalone unit which only requires power and GPS antenna connection. The full schematic and block diagram of the synthesizer chipset is presented along with measured output waveform, frequency an phase noise at 1GHz. You can purchase your own GPS DO here.
33:28 – Sain Smart USB-C TS80 Portable Soldering Iron
This soldering iron which supports QC 3.0 is the followup of the very popular TS100 soldering iron. The unit can operate with up to 18W of power from 9V. However, it does not support PD on USB-C. By using the AVHzY USB tester the power bank can be configured to provide any desired output voltage and deliver up to 28W into the TS80 soldering iron. Using this method, the TS80 soldering iron can easily solder large components on a wide ground plane. The TS80 soldering iron can be purchased here or internationally from here. The Ankar PD+ 28600mAh power bank can be purchased from here and the AVHzY USB tester can be purchased here.
A laptop for Macy who is about to start college
New shoes for Emily who loves to walk around campus
New Clothes for Myla who loves to play outside
A new pair of school shoes for Lemon who loves all sports
A new digital watch to remind Kevin to take his medicine
A new coat for Thad to stay warm and look cool
New clothes for Sharain who is looking for new employment after graduation
New work clothes for Christine who is excited to start her new job
A new Desk Lamp for Cristal so she can focus on homework
Sensory tools for Wyatt who needs a bit of extra help
Safe transportation for Shakeny who loves musica
New clothes for Isaiah who is very clever
Some new clothes for Katelyn who is making a change in her life
A new pair of sneakers for Sohelia the avid dog walker
The Signal Path (TSP) is an electrical engineering video blog for industry professionals, students and hobbyists. TSP is a non-for-profit website dedicated to provide free education spanning a wide range of electrical engineering topics. Equipment reviews, tutorials and repair videos are posted regularly.