In this episode Shahriar reviews the top-of-the-line Tektronix 6-Series Oscilloscope. This 4-channel instrument offers 8-GHz of bandwidth at 25GS/s on all channel independently. Tektronix has made great strides in offering low-noise front-end custom ASICs combined with hardware digital down-conversion built into the core of the 6-Series. This enables advanced triggering across in both time and frequency domains as well as multi-domain correlated measurements.
This review is organized as follows:
00:06 – Introductions
01:42 – Instrument design, front and back panels
03:34 – Full acquisition board teardown, analysis and architecture
18:11 – PLL Experiment: Spectrum View, advanced triggering, PLL characterization and debugging
38:33 – Backplane Communication: Jitter analysis, eye diagrams, jitter composition, cross-talk
49:24 – Concluding remarks
In this episode Shahriar investigates the failure of a Voltech PM3000A 3-Phase Universal Power Analyzer. This instrument is capable of advanced measurements on power profile and performance of a DUT. The unit shows “Voltage & Current OVERLOAD” condition on all channels even without a DUT connected.
The teardown of the unit reveals the isolation techniques used between the analog sampling heads and the main digital/DSP board. The instrument uses a combination of transformers and optocouplers to provide power to the sampling heads and receive digitized data from the voltage/current ADCs.
Close examination of the unit shows a faulty capacitor on the AC power supply feeding the sampling heads. The capacitor is replaced which corrects the overload condition. The unit is then used to measure the power profile of a simple glue-gun. The PF and THD measurements show that the glue-gun is equipped with a triac for power control; an expected result for a simple glue-gun.
In this episode Shahriar troubleshoots a defective Agilent N9020A 8.4GHz spectrum analyzer. The instrument reports LO Unlock and refuses to perform any sweep at all. The complete block diagram of the instrument is presented with focus on the LO generation blocks.
The teardown of the instrument reveals the individual assemblies presented during the block diagram overview. After a few measurements, it is concluded that the LO Synthesizer module is faulty. The module is presented in detail and various components and signal paths are identified. Extensive measurements are carried out until the fault is traced to problems in the varactor bank of the main VCO. After the component replacement, the instrument is fully functional. The X-Ray of the synthesizer module is also presented during the repair.
In this episode Shahriar reviews the new Tektronix AFG31000 Arbitrary Function Generator. The AFG31000 series with InstaView™ technology is the first high-performance AFG with built-in waveform generation applications, patented real-time wave monitoring, and a modern user interface. The instrument is capable of 2GS/s waveform generation up to 250MHz and 14-bits of vertical resolution. This review is organized as follows:
00:08 – Introductions
01:01 – Front panel, rear panel and instrument design
03:44 – Detailed teardown, analysis, architecture and components
17:30 – GUI overview and basic functions
20:25 – Broadband FM modulation experiments
23:55 – InstaView experiments with a non-linear impedance
35:38 – Equation editor, waveform builder and advance sequencing
51:21 – Concluding remarks
In this episode Shahriar takes investigates the architecture and benefits of a precision sub-sampling wide-bandwidth oscilloscope. The teardown of the Agilent DCA instrument reveals a PC core combined with the acquisition board. The instrument without any plug-in modules does not perform any exciting tasks.
A 20GHz sampling module from eBay is available. However, the unit does not have an front connectors. New connectors are sourced from an old timing module and the sampling head is back to working condition. In order to test the unit, an Optellent OBP-04×10 4-channel pattern generator and error counter is used. The teardown of the unit reveals a simple design built around several transceiver ICs. The comparison between the sub-sampling scope and a real-time scope is also presented alongside measured results from the DCA.
The Signal Path (TSP) is an electrical engineering video blog for industry professionals, students and hobbyists. TSP is a non-for-profit website dedicated to provide free education spanning a wide range of electrical engineering topics. Equipment reviews, tutorials and repair videos are posted regularly.