In this episode Shahriar diagnoses a malfunctioning Agilent 86118A 70GHz Dual Remote Sampling Head. With 70GHz of analog bandwidth per channel, this module definitely deserves a repair.
Unfortunately most failed remote-head modules aren’t repairable if the damage is in the sensitive proprietary III/V front-end IC. However, in this situation the module intermittently fails calibration which is more promising. The teardown reveals the architecture of the module from the point of view of the analog and strobe paths. Measurements on the strobe path also show a potential failure on the power-amplifier/splitter network which feeds the two remote-heads. A pair of transistors are replaced on the strobe path which correct the problem. The unit passes calibration and is then used for various eye-diagram measurements to verify full functionality.