In this episode Shahriar takes a look at a non-functional R&S FSEK 40GHz spectrum analyzer. The instrument’s LCD screen back light no longer works and the front glass is broken. Furthermore, no testing can be done until these problems are resolved. The CCFL tube is replaced with an LED counterpart and a new front panel glass is also installed.
After power on it becomes clear that the instrument has other defects. The internal PLL is unlocked. It is observed that the OCXO module is cold and potentially not working. Examination of the LO board shows a bad inductor on the DC filter side. After this problem the instrument operates correctly but only for signals below 7GHz. After further testing it is discovered that the YIG tuned filter on the pre-selector path is not tuning at all. Eventually the fault is traced to some operational amplifiers on the YIG driver board. The repaired instrument is verified for correct operation.