In this episode Shahriar investigates a malfunctioning E3646A Dual Channel Programmable Power Supply. The power supply provides the correct output voltage on one channel. However, the second channel produces no output and continuously reads 0V and 0A.
Applying an external voltage to the malfunctioning channel produces the correct readout from the instrument display suggesting that most of the channel circuitry is functional. After a close examination of the power supply schematics, the problem is traced to a failed solder joint. The DAC output amplifier is disconnected from the output transistor and after repairing the colder joint, the power supply is fully functional.