In this episode Shahriar continues his review of the Tektronix 5-Series 8-Channel MSO Oscilloscope following the original overview of the instrument. A full teardown of the instrument as well as a set of experiments showcasing the new Spectrum View capability are shown. This review is organized as follows:
00:09 – Introduction and outline
01:31 – Teardown and analysis of the acquisition board
13:38 – Experiment setup and description
16:59 – Measurement results and Spectrum View capabilities
33:12 – Concluding remarks
In this episode Shahriar reviews the very unique SAF Spectrum Compact V-Band portable spectrum analyzer. These battery powered devices do not require external down-converters, and a particularly useful tool for any field engineer performing microwave equipment installation, link troubleshooting, site planning and maintenance. The analyzers utilize a resistive touch screen, allowing the engineer to wear gloves when using the device. The Spectrum Compact can be used for high precision detection of existing interference on installed paths or available radio channels. Data logging of all spectrum scans is available with the Spectrum Compact device, and enhanced data processing and analysis are provided by SAF Tehnika designed PC software for your laptop.
The instrument teardown reveals a multi-board design and each module’s functionality and design is presented. The unit is then used in several experiments to track and analyze various V-Band RF signals. The review is organized as follows:
01:00 – Overview and model comparison
03:16 – Instrument design, overview power-on benefits
05:23 – Instrument GUI overview
09:56 – Full teardown and analysis of all modules
19:22 – Experiments with various unknown V-Band RF signals
29:43 – PC software overview and capabilities
31:30 – Concluding remarks
In this episode Shahriar repairs an Agilent N9020A MAX Spectrum Analyzer. The instrument is well equipped with real-time capability and direct I/Q input ports on the front panel. The instrument does not show the correct signal levels and fails internal calibration.
After a brief teardown, the block diagram of the instrument is carefully examined for potential failure points. It is determined that front-end switch and mechanical attenuator is likely faulty. Teardown of the unit reveals significant damage to the attenuator due to some high-power at the input. Unfortunately, this also implies that the front-end RF selector and pre-amplifier modules are also damaged. In the second repair video, the front-end module is disassembled and damage to the dies can be observed.
Luckily, a relatively inexpensive front-end (SSLAM) unit is sourced from eBay and the repair is completed. The performance of the unit is verified after the repair and the front-end I/Q input functionality is demonstrated.
In this episode Shahriar reviews three mailbag items. There is also a new giveaway! Follow the instruction at the beginning of the video to enter. The reviews are organized at follows:
00:09 – Introduction & Giveaway Details
02:21 – Leo Bodnar Electronic – Fast Pulse Generator
These fast pulse generators are rated to below 25ps! The modules are examined and various microwave and design considerations are explored. The block diagram and functionality of the final output driver is also described in details. The units are used to measure the rise and fall time of various oscilloscopes ranging from 300MHz to 13GHz of analog bandwidth. You can purchase your own fast pulse generator here.
21:11 – SV1AFN GPS Disciplined Oscillator
This GPS disciplined oscillator not only provides frequency synchronization to the GPS clock, it also provides programmable outputs on two independent channels using a Silicon Labs synthesizer IC. The complete module acts as standalone unit which only requires power and GPS antenna connection. The full schematic and block diagram of the synthesizer chipset is presented along with measured output waveform, frequency an phase noise at 1GHz. You can purchase your own GPS DO here.
33:28 – Sain Smart USB-C TS80 Portable Soldering Iron
This soldering iron which supports QC 3.0 is the followup of the very popular TS100 soldering iron. The unit can operate with up to 18W of power from 9V. However, it does not support PD on USB-C. By using the AVHzY USB tester the power bank can be configured to provide any desired output voltage and deliver up to 28W into the TS80 soldering iron. Using this method, the TS80 soldering iron can easily solder large components on a wide ground plane. The TS80 soldering iron can be purchased here or internationally from here. The Ankar PD+ 28600mAh power bank can be purchased from here and the AVHzY USB tester can be purchased here.
A laptop for Macy who is about to start college
New shoes for Emily who loves to walk around campus
New Clothes for Myla who loves to play outside
A new pair of school shoes for Lemon who loves all sports
A new digital watch to remind Kevin to take his medicine
A new coat for Thad to stay warm and look cool
New clothes for Sharain who is looking for new employment after graduation
New work clothes for Christine who is excited to start her new job
A new Desk Lamp for Cristal so she can focus on homework
Sensory tools for Wyatt who needs a bit of extra help
Safe transportation for Shakeny who loves musica
New clothes for Isaiah who is very clever
Some new clothes for Katelyn who is making a change in her life
A new pair of sneakers for Sohelia the avid dog walker
In this episode Shahriar investigates the failure of an Anritsu MS2712B 1.7GHz portable spectrum analyzer with built-in tracking generator. The instrument is missing the main firmware flash card which is easily replaced. A full teardown of the instrument is presented with focus on both the tracking generator PCB and main RF deck. Various ASICs and microwave components are closely examined. To help with removing and replacing many microwave screws, a Hitachi/Metabo electric screwdriver is used which can be purchased here.
After re-assembly of the unit, it becomes apparent that a mechanical push on the RF deck causes the LO1 PLL to unlock. This points to a fault in one of the microwave components. The issue is traced to a broken solder joint inside of the VCO modules. After repair, the instrument functions normally. The performance and tracking generator functionality is verified by measuring a tuneable band-pass filter.
The Signal Path (TSP) is an electrical engineering video blog for industry professionals, students and hobbyists. TSP is a non-for-profit website dedicated to provide free education spanning a wide range of electrical engineering topics. Equipment reviews, tutorials and repair videos are posted regularly.