In this episode Shahriar repairs an Agilent PSA Series Spectrum Analyzer. The instrument generates many errors during self-alignment and produces no measurements below 3.2GHz. The block diagram of the unit is thoroughly presented and various possible failure points are considered. Based on the observation of the noise floor, the most likely cause is the second LO module. The measurement of the LO power indicates that the second LO power is fall below nominal.
The second LO signal (at 3.6GHz) is generated by a DRO PLL module which is locked to 600MHz. The PLL is functional, however the output power is below -20dBm. Teardown of the module reveals a simple design with a doubler and filter. The filter (which is discolored) shows a huge loss likely caused by RF losses due to trace degradation. Thinning the traces provides some improvement and the remaining loss is compensated with a two stage RF amplifier. With this modifications, the instrument’s alignment errors are partially resolved.
The next problem is with the input attenuator at the 20dB range. This problem is simply resolved by providing lubrication on the attenuator solenoids after disassembly. The instrument is then used for various measurements to verify its correct operation.
In this episode Shahriar investigates the noise generated from the DPH3205 Buck-Boost DC/DC Converter Module. This module is part of a family of affordable devices which can be purchased online. The particular model (DPH3205) shown in this video can be purchased from here. The Keysight S-Scope in conjunction with the N7020 power probe is used to measure the noise behavior of the circuit. It can be observed that the module produces many harmonics of noise spanning up to and beyond 1MHz.
A power operational amplifier is used in conjunction with a 4V Zener diode to produce a crude voltage follower. The PSRR of the operational amplifier can filter the noise from the DC/DC converter. This is verified by measuring the noise coming from the operational amplifier. The noise is significantly reduced. This circuit is not entirely practical and is used for demonstration purposes only.
In this episode Shahriar explores the cause of failure of an Agilent E4405B ESA 13.5GHz spectrum analyzer. In a previous episode, an E4407B version of the ESA was repaired and upgraded which is helpful in debugging and analyzing this instrument. The instrument displays LO Unlock which is a familiar problem. After an overview of the system block diagram, the LO signal path is traced and analyzed for problems.
After some investigation, the problem appears to be identical to the E4407B model repaired before! The Hittite (Analog Devices) divide-by-4 static divider unit has failed and does not divide correctly at high frequencies. The component is replaced which restores the instrument’s functionality.
The LCD screen of the unit is also very dim. The CCFL tube and the inverter are replaced with an LED back light instead. The unit then produces a bright and vibrant graphs and text. The performance of the instrument is verified with a Keysight MXG signal generator.
In this episode Shahriar & Rosanah investigate an Agilent power supply which does not appear to power on. It can be quickly observed that the fuse has failed on the unit. Using an isolation transformer a small amount of AC voltage is applied to the unit after the fuse replacement. It is clear that a short is present somewhere in the instrument since even at 10V AC the instrument consumes more than 1A.
After separating the internal transformer from the main PCB, the high current consumption is eliminated. The short is traced to a damaged main rectifier in the instrument. A replacement rectifier is used which allows the instrument to power on.
The VFD display is however extremely dim. After a failed attempt at restoring the VFD brightness, a new display is purchased from eBay. The replacement produces a bright vibrant VFD display. The performance of the power supply is verified under load.
In this episode Shahriar repair a Fluke PM6685R Rubidium Frequency Counter. The instrument shows highly unstable results and produces high-pitch noise when powered on. The high-frequency noises point to a problem with the DC-DC converter circuit in the power supply module. A close examination points to degraded capacitors. After all capacitors have been replaced, the noise issue is resolved. The instrument’s performance is then verified and calibrated against an external Rubidium reference. The agreement between the units is better than 10ppb.
The Signal Path (TSP) is an electrical engineering video blog for industry professionals, students and hobbyists. TSP is a non-for-profit website dedicated to provide free education spanning a wide range of electrical engineering topics. Equipment reviews, tutorials and repair videos are posted regularly.