In this episode Shahriar repair a Fluke PM6685R Rubidium Frequency Counter. The instrument shows highly unstable results and produces high-pitch noise when powered on. The high-frequency noises point to a problem with the DC-DC converter circuit in the power supply module. A close examination points to degraded capacitors. After all capacitors have been replaced, the noise issue is resolved. The instrument’s performance is then verified and calibrated against an external Rubidium reference. The agreement between the units is better than 10ppb.
Tag Archive for Analog
In this episode Shahriar investigates a peculiar problem with a SRS DS345 function generator. The instrument does not appear to produce the correct output signal frequency. After some investigation it becomes clear that a few of the display digits are not active which hide the actual frequency settings. The problem is traced to a broken resistor on the display driver. The schematic of the instrument shows that the resistor is responsible for strobing the affected seven-segment digit. After the repair, the instrument’s performance is verified including the OCXO accuracy.
In this episode Shahriar investigates a faulty Agilent 53152A 46GHz frequency counter. The instrument does not power on and shows no sign of internal voltage presence. Teardown of the instrument reveals a large PCB where all analog and digital circuity is contained. The power supply module is a module components and upon measurements shows no activity.
The power supply is a simple switching architecture with functioning input rectifier and capacitor filter. By using an oscilloscope it is clear that the power supply PWM controller attempts to start. However, the main power supply pin shows unstable voltages indicating inadequate charge retention on the rectifying capacitor. Replacing the capacitor revives the startup condition and the power supply function returns. The PWM controller and main switching transistors are also replaced with new ones. After this repair the unit powers on and passes all self-tests. The unit can successfully measure signal frequencies and power.
In this short episode Shahriar demonstrates an overview of the new Tektronix MSO58 8-Channel 6.25GS/s 2GHz Mixed-Signal Oscilloscope with up to 64-Bits of digital channels. The scope features independent ADC, data acquisition and memory per channel as well as FlexChannel architecture which allows each channel to act as either a 2GHz analog channel or an 8-Bit digital channel. The scope also features a brand new GUI interface on an HD 15.6″ display with significant performance enhancement and touch interface optimizations. A full review of the instrument will be provided in the future.
In this short episode Shahriar repairs a more affordable instrument! This BK Precision LCR meter does not power up despite looking almost new. The teardown of the unit reveals several damaged components. Without any schematic and obscure part numbers some circuit configurations have to be reverse engineered. One of the main failed components is a TI DC-DC converter IC whose datahseet is examined during the repair.
The damaged components are replaced and the unit powers on normally. The performance of the instrument is verified using a HP LCR meter. The demonstrated Soldering Iron Stand can be found here.