In this episode Shahriar investigates a faulty Agilent 53152A 46GHz frequency counter. The instrument does not power on and shows no sign of internal voltage presence. Teardown of the instrument reveals a large PCB where all analog and digital circuity is contained. The power supply module is a module components and upon measurements shows no activity.
The power supply is a simple switching architecture with functioning input rectifier and capacitor filter. By using an oscilloscope it is clear that the power supply PWM controller attempts to start. However, the main power supply pin shows unstable voltages indicating inadequate charge retention on the rectifying capacitor. Replacing the capacitor revives the startup condition and the power supply function returns. The PWM controller and main switching transistors are also replaced with new ones. After this repair the unit powers on and passes all self-tests. The unit can successfully measure signal frequencies and power.
In this short episode Shahriar demonstrates an overview of the new Tektronix MSO58 8-Channel 6.25GS/s 2GHz Mixed-Signal Oscilloscope with up to 64-Bits of digital channels. The scope features independent ADC, data acquisition and memory per channel as well as FlexChannel architecture which allows each channel to act as either a 2GHz analog channel or an 8-Bit digital channel. The scope also features a brand new GUI interface on an HD 15.6″ display with significant performance enhancement and touch interface optimizations. A full review of the instrument will be provided in the future.
In this short episode Shahriar repairs a more affordable instrument! This BK Precision LCR meter does not power up despite looking almost new. The teardown of the unit reveals several damaged components. Without any schematic and obscure part numbers some circuit configurations have to be reverse engineered. One of the main failed components is a TI DC-DC converter IC whose datahseet is examined during the repair.
The damaged components are replaced and the unit powers on normally. The performance of the instrument is verified using a HP LCR meter. The demonstrated Soldering Iron Stand can be found here.
The existing decade resistance box shows inconsistent measurements and intermittent contacts prior to the DeoxIT application. After spraying all contacts the instrument is significantly improved with repeatable and accurate resistance measurements. DeoxIT can be purchased from a variety of suppliers and vendors including Amazon.
In this episode Shahriar attempts a difficult repair of a Tektronix RSA real-time spectrum analyzer. This well-equipped instrument reports several error messages during startup POST including LO Unlock as well as Signal Path failures. The service manual of the instrument does not provide any detailed block diagram and no schematics. Most failures require the instrument to be serviced by the Tektronix factory. The equipment has various advanced options including 110MHz analysis bandwidth, digital modulation analysis, wide-band IF output and deep memory.
The teardown of the instrument reveals a multi-board, multi-module design. The top of the instrument contains all digital blocks and boards while the bottom of the instrument houses the RF deck assembly. The RF deck is broken into various stages such as attenuator, RF switches, first converter, second converter, IF block, reference synthesizer and LO generator. For the purposes of addressing the YIG problems the LO board is examined. The problem is traced to two components, both dividers in the complex PLL system of the LO subsystem. The LO board is fully analyzed and described and the defective components re replaced.
The repaired instrument is tested for various functionality including DPX, de-modulation and measurement of beyond 8GHz CW tones. The instrument passes all self-tests, alignments and detailed diagnostics.
The Signal Path (TSP) is an electrical engineering video blog for industry professionals, students and hobbyists. TSP is a non-for-profit website dedicated to provide free education spanning a wide range of electrical engineering topics. Equipment reviews, tutorials and repair videos are posted regularly.