In this episode Shahriar reviews three mailbag items. There is also a new giveaway! Follow the instruction at the beginning of the video to enter. The reviews are organized at follows:
00:09 – Introduction & Giveaway Details
02:21 – Leo Bodnar Electronic – Fast Pulse Generator
These fast pulse generators are rated to below 25ps! The modules are examined and various microwave and design considerations are explored. The block diagram and functionality of the final output driver is also described in details. The units are used to measure the rise and fall time of various oscilloscopes ranging from 300MHz to 13GHz of analog bandwidth. You can purchase your own fast pulse generator here.
21:11 – SV1AFN GPS Disciplined Oscillator
This GPS disciplined oscillator not only provides frequency synchronization to the GPS clock, it also provides programmable outputs on two independent channels using a Silicon Labs synthesizer IC. The complete module acts as standalone unit which only requires power and GPS antenna connection. The full schematic and block diagram of the synthesizer chipset is presented along with measured output waveform, frequency an phase noise at 1GHz. You can purchase your own GPS DO here.
33:28 – Sain Smart USB-C TS80 Portable Soldering Iron
This soldering iron which supports QC 3.0 is the followup of the very popular TS100 soldering iron. The unit can operate with up to 18W of power from 9V. However, it does not support PD on USB-C. By using the AVHzY USB tester the power bank can be configured to provide any desired output voltage and deliver up to 28W into the TS80 soldering iron. Using this method, the TS80 soldering iron can easily solder large components on a wide ground plane. The TS80 soldering iron can be purchased here or internationally from here. The Ankar PD+ 28600mAh power bank can be purchased from here and the AVHzY USB tester can be purchased here.
A laptop for Macy who is about to start college – $260
New shoes for Emily who loves to walk around campus – $55
New Clothes for Myla who loves to play outside – $55
A new pair of school shoes for Lemon who loves all sports – $55
A new digital watch to remind Kevin to take his medicine – $74
A new coat for Thad to stay warm and look cool – $47
New clothes for Sharain who is looking for new employment after graduation – $55
New work clothes for Christine who is excited to start her new job – $55
A new Desk Lamp for Cristal so she can focus on homework – $55
Sensory tools for Wyatt who needs a bit of extra help – $111
Safe transportation for Shakeny who loves musica – $55
New clothes for Isaiah who is very clever – $55
Some new clothes for Katelyn who is making a change in her life. – $55
A new pair of sneakers for Sohelia the avid dog walker – $86
In this episode Shahriar takes a look at one of the most advanced electrical test and measurement instruments ever created. The Keysight UXR-Series Real-Time Oscilloscope brings 110GHz of analog bandwidth and 256GS/s real-time sampling at 4-channels simultaneously. To make it even more impressive, the entire data-conversion architecture is in 10-bits. This implies that the instruments captures, processes, stores and displays over 10Tb/s of information.
Various architectures of state-of-the art oscilloscopes from Keysight, LeCroy and Tektronix are examined and compared against the new real-time architecture of the UXR-Series oscilloscope. The teardown of the front-end 110GHz module along with the data acquisition board is presented and analyzed in detail. The instrument showcases a wide range of Keysight technologies implemented in various technologies such as InP, SiGe BiCMOS, 65nm CMOS and 28nm CMOS nodes. In combination with Hyper-Cube memory module, data can be captured at 256GS/S from all 4-channels at the same time. Several variants of the UXR-Series oscilloscope will be available from 13GHz to 110GHz bandwidths.
A new calibration probe is also introduced based on the Keysight InP process capable of producing signal edges with sub-3.5ps of rise/fall times with NIST traceable calibration data. This enables users to perform NIST alignment and bandwidth calibration on site without needing to send the instrument back to Keysight.
Several measurements with the scope demonstrates its extraordinarily low noise floor, jitter as well as the capability of the new probe module for instrument calibration. The 110GHz 4-channel variant of the UXR-Series oscilloscope has an MSRP of $1.3 Million US dollars.
In this episode Shahriar repairs an Agilent 4338B milliohmmeter. This instrument is capable of measuring extremely small resistances down to 10uΩ while maintaining a DUT voltage of less than 20mV. The instrument powers on with the message ADC Failure. Investigation reveals that the instrument uses an obsolete ADC which must be removed from the board in order to reverse engineer its operation. While the ADC turns out to be functional, a PAL device which controls the ADC timing is faulty. A new device is salvaged from a donor board to complete the repair. The instrument is then used to measure several small known resistances.
In this episode Shahriar investigates an Agilent E4981A high-speed capacitance meter which fails self-test during startup. The instrument reports a series of errors associated with the CPU board. The teardown reveals a multi-board construction where the main analog assembly employs an FPGA for control and interface to the main CPU board. The unit has a disconnected ribbon cable between the analog board and the main CPU assembly. After correcting the problem the instrument completes self-test without errors.
After the repair the unit is used to measure several known capacitors for functionality verification. Finally the instrument is used to measure a hand-made parallel plate capacitor. The capacitor can accept various types of dielectric material whose impact is studied.
The Signal Path (TSP) is an electrical engineering video blog for industry professionals, students and hobbyists. TSP is a non-for-profit website dedicated to provide free education spanning a wide range of electrical engineering topics. Equipment reviews, tutorials and repair videos are posted regularly.