In this episode Shahriar reviews the long awaited Keysight MXA Signal Analyzer (N9020B). The new X-Series Spectrum Analyzers from Keysight offer an entirely re-designed GUI interface which supports multiple tabs as well as multi-touch interaction. The review is organized as follows:
1:07 – Model comparison and overview of x-series spectrum analyzers.
7:08 – Instrument physical overview.
11:49 – New GUI overview and demonstrations.
22:44 – Instrument block diagrams and analysis.
28:46 – Brief look inside the MXA spectrum analyzer.
32:53 – Back-scattering communication experiment block diagram and description.
37:16 – Phase noise characterization of doubler and amplifier.
51:54 – Complete back-scattering experiment measurements and analysis.
1:04:45 – Analysis of broadband modulated signals by using the S-Series scope and MXA Signal analyzer at the same time.
1:13:56 – Concluding remarks.
The existing decade resistance box shows inconsistent measurements and intermittent contacts prior to the DeoxIT application. After spraying all contacts the instrument is significantly improved with repeatable and accurate resistance measurements. DeoxIT can be purchased from a variety of suppliers and vendors including Amazon.
In this short episode Shahriar takes a detailed look at an Anritsu variable optical attenuator operating in the L- and C-Bands. The mechanical attenuator is defective and provides an opportunity for complete disassembly and examination of the free-space optics, mechanical components and electronics.
The electronics drive a pair of DC motors with potentiometer decoders in feedback along with operational amplifiers and a DAC. This allows the microprocessor to set the exact position of the motors based on calibrated values stored on EPROM. The free-space optics consists of two back-to-back disks with radial metal coatings. Depending on the position of each wheel, the coating limits the transmission of the light through the wheel causing attenuation. The entire housing is hermetically sealed to prevent degradation of the disks.
In this episode Shahriar attempts a difficult repair of a Tektronix RSA real-time spectrum analyzer. This well-equipped instrument reports several error messages during startup POST including LO Unlock as well as Signal Path failures. The service manual of the instrument does not provide any detailed block diagram and no schematics. Most failures require the instrument to be serviced by the Tektronix factory. The equipment has various advanced options including 110MHz analysis bandwidth, digital modulation analysis, wide-band IF output and deep memory.
The teardown of the instrument reveals a multi-board, multi-module design. The top of the instrument contains all digital blocks and boards while the bottom of the instrument houses the RF deck assembly. The RF deck is broken into various stages such as attenuator, RF switches, first converter, second converter, IF block, reference synthesizer and LO generator. For the purposes of addressing the YIG problems the LO board is examined. The problem is traced to two components, both dividers in the complex PLL system of the LO subsystem. The LO board is fully analyzed and described and the defective components re replaced.
The repaired instrument is tested for various functionality including DPX, de-modulation and measurement of beyond 8GHz CW tones. The instrument passes all self-tests, alignments and detailed diagnostics.
In this short episode Shahriar takes a close look at an Ist-Rees laser spectrum analyzer. This simple instrument is based on a continuously rotating diffraction mirror to detect various wavelengths. A trigger signal is used to denote the beginning of the scan and the front-panel display shows the current selected wavelength. By aligning the output signal with the trigger signal the wavelength of the input light can be measured.
After the unit teardown, the instrument is used to measure the wavelength of a semiconductor 1310nm laser. The unit is calibrated using this source and is then used to measure the infrared leakage wavelength from a green laser pointer.
The Signal Path (TSP) is an electrical engineering video blog for industry professionals, students and hobbyists. TSP is a non-for-profit website dedicated to provide free education spanning a wide range of electrical engineering topics. Equipment reviews, tutorials and repair videos are posted regularly.