In this episode Shahriar demonstrates the functionality and applications of an Agilent 11896A Polarization Controller. Various fiber optic communication methods are presented. This includes the use of complex modulation schemes (such as PAM and QAM for coherent receivers), polarization division multiplexing (PDM), wavelength division multiplexing (WDM), and spatial division multiplexing (SDM). The concept of light polarization is demonstrated by using a pair lenses from a consumer theater 3D glasses and two blue LEDs with uncorrelated lighting patterns.
In order to test the polarization controller, a solid-state laser source, SMF fiber with APC/PC connectors as well as a polarization beam splitter is presented. By using a pair of optical power sensors, the functionality of the polarization controller is verified. Finally, the teardown of the unit is presented and the method to achieve polarization control is observed.
The fiber optic communication overview document can be downloaded here. I’d also like to acknowledge my colleague and friend Dr. Timo Pfau for his expertise and consult on fiber optic communication methods.
In this episode Shahriar goes through the teardown and repair process of a Fluke 196B Handheld ScopeMeter. This unit, which has been purchased as a “dead” unit from eBay, is examined for faults. During the teardown, various PCB components, architecture and design ispresented. The performance and functionality of the unit is verified with a few brief experiments. Finally, the schematic and block diagram of the Fluke 196B is explored for the finer details of its design and operation.
In this episode Shahriar attempts to repair an HP 8648C Synthesized Signal Generator from a dumpster dive. The cause of failure is investigated and traced. Alongside the repair efforts, the internal construction and components of the synthesizer are also presented. Various component teardowns are shown and described including the structure of the VCO used in the generator assembly. The functionality of the repaired unit is verified using an Agilent Power Meter and Rigol Spectrum Analyzer. Finally, the overall unit block diagram is presented and the principle of operation is described in detail.
In this episode Shahriar presents the teardown of an Advantest TQ8325 Wavelength Meter. The principle operation of the instrument is explained along with a close up of the free space optics in operation. To verify the functionality of the unit, an infrared laser module is used as a CW coherent source. The impact of the laser diode’s temperature variation on the output wavelength is measured using the wavelength meter.
I’d like to Acknowledge Dr. Jeffery Lee for his consult and expertise on free space optics and fiber optic measurement systems.
In this episode Shahriar takes a look inside the “Logic”, “Logic16″ and USBee SX which were reviewed in the previous episode. The PCBs get a closeup inspection along with some insight into the design of these products. How do they compare? Well, you’ll have to watch the video to find out! Make sure to comment and join our Community Forum to discuss these products.
This is "The Signal Path dot com"! An electrical engineering video blog for students, hobbyists and hackers. I do equipment reviews, teardowns, circuit design tutorials, testing procedures and more! Subscribe to my RSS, join the community forum and participate in the discussions.