In this two-part episode Shahriar repairs an Agilent N5230A 13.5GHz PNA-L which suffers from un-leveled output in full frequency sweep range. The instrument is not able to produce any outputs above -25dBm above 10.5GHz and shows below specification output power levels below 10.5GHz.
Teardown of the unit reveals various microwave modules with the SSLAM final module responsible for amplitude leveling and frequency multiplication for frequencies above 10.5GHz. Teardown of this module shows a collection of MMIC dies wirebonded together using micro-strip ceramic pieces and interconnects. MMIC include switches, amplifiers, filters, multipliers, power detectors and attenuators. After examining the input/output signals of the module it becomes clear that the input switch IC is damaged. Unfortunately, an attempt to replace the die with a packaged QFN GaAs device fails. However, a lucky eBay find of a used SSLAM module saves the day and the instrument is restored to full functionality. Some measurements of an packaged evaluation board of a GaAs IC is demonstrated using the newly repaired PNA-L.
In this episode Shahriar analyzes the failure of two Agilent 8449B Preamplifiers. These units should provide up to 30dBm of gain from 1.0GHz to 26.5GHz intended for use as a preamplifier. Both amplifiers test positive for power supply voltages and operation. After removing the interface cables from the amplifier module to the front panel, it becomes clear that mechanical shock has caused damage to the front panel connectors. Replacement cables and connectors are used to correct the problem.
In this episode Shahriar repairs an Agilent 4338B milliohmmeter. This instrument is capable of measuring extremely small resistances down to 10uΩ while maintaining a DUT voltage of less than 20mV. The instrument powers on with the message ADC Failure. Investigation reveals that the instrument uses an obsolete ADC which must be removed from the board in order to reverse engineer its operation. While the ADC turns out to be functional, a PAL device which controls the ADC timing is faulty. A new device is salvaged from a donor board to complete the repair. The instrument is then used to measure several small known resistances.
In this episode Shahriar investigates an Agilent E4981A high-speed capacitance meter which fails self-test during startup. The instrument reports a series of errors associated with the CPU board. The teardown reveals a multi-board construction where the main analog assembly employs an FPGA for control and interface to the main CPU board. The unit has a disconnected ribbon cable between the analog board and the main CPU assembly. After correcting the problem the instrument completes self-test without errors.
After the repair the unit is used to measure several known capacitors for functionality verification. Finally the instrument is used to measure a hand-made parallel plate capacitor. The capacitor can accept various types of dielectric material whose impact is studied.
In this episode Shahriar takes a close look at a defective Agilent E4440A spectrum analyzer. Initial analysis reveals that the startup errors are caused by a defective pre-amplifier module. During initial alignment and calibration, the defective microwave relays inside the pre-amplifier module fail insertion loss specifications and cause alignment failures.
Unfortunately, the relays are not repairable and must be replaced. Two new relays are purchased and replacement clears all alignment issues. The main motherboard battery has also failed; a replacement battery corrects any time and date keeping errors. The repaired instrument is cycled through several alignment procedures without any problems. The performance of the unit is verified by examining the DANL of the instrument with and without the pre-amplifier activated.
The Signal Path (TSP) is an electrical engineering video blog for industry professionals, students and hobbyists. TSP is a non-for-profit website dedicated to provide free education spanning a wide range of electrical engineering topics. Equipment reviews, tutorials and repair videos are posted regularly.