In this episode Shahriar troubleshoots a defective Agilent N9020A 8.4GHz spectrum analyzer. The instrument reports LO Unlock and refuses to perform any sweep at all. The complete block diagram of the instrument is presented with focus on the LO generation blocks.
The teardown of the instrument reveals the individual assemblies presented during the block diagram overview. After a few measurements, it is concluded that the LO Synthesizer module is faulty. The module is presented in detail and various components and signal paths are identified. Extensive measurements are carried out until the fault is traced to problems in the varactor bank of the main VCO. After the component replacement, the instrument is fully functional. The X-Ray of the synthesizer module is also presented during the repair.