In this episode Shahriar takes a close look at a faulty Agilent E4433B 4.0GHz signal generator and vector modulator. This instrument is equipped with many of the most useful options such as dual arbitrary waveform generator and high-power output.
The instrument does not generate any error messages, however the output signal is ~20dB below specifications. After presenting the system block diagram and observing the output power characteristics, it becomes clear that the issue is with the mechanical attenuator. The teardown of the attenuator reveals faulty o-rings on the 5dB attenuation pad which has to be repaired with some flexible epoxy.
After the repair the instrument’s functionality is verified by measuring the output power with a power meter, frequency accuracy with a Rubidium frequency counter and the modulation accuracy with a vector signal analyzer.
Hitachi/Metabo electric can be purchased here.