In this episode Shahriar takes a close look at an EG&G DSP-Based Lock-In Amplifier. This instrument provides excellent sensitivity for both voltage and current inputs and is based on a fully digital architecture. The instrument’s LCD screen is dim and unreadable. Furthermore, a few lines are missing from the LCD pointing to a failed zebra strip.
The instrument’s block diagram is examined and explained. The teardown reveals the structure of the LCD screen, front-end blocks and internal PLL. The CCFL back-light inverter is located and replaced with a low-voltage DC-DC converter for use with an LED strip. The instrument’s functionality is verified through a few experiments including measuring the 3dB bandwidth of an RC filter as well as measuring the photo-current induced in an LED using a synchronized infrared light source.