In this episode Shahriar repairs an Agilent 54845A oscilloscope with an intermittent failure. The instrument occasionally fails self-calibration and the displayed waveform shows undesired spurious activities. The teardown of the instrument reveals customized front-end section coupled into a ceramic-substrate ADC ASIC designed by Agilent. The packaged ADCs are placed in sockets. The ICs are removed and the ceramic carrier is cleaned; this resolves the intermittent problem with the channels. The architecture of the scope is also presented. The functionality of the scope is verified through self-calibration, self-test and measurement of various waveform.