In this episode Shahriar examines a faulty R&S FSH3 100kHz – 3.0GHz Portable Spectrum Analyzer. This exceptionally dirty unit does not power on and is missing a power supply and battery. After verifying the correct operation of the battery charger circuit and keypad the problem is traces to a damaged SOT-23-6 MOSFET device which is part of a fly-back DC-DC converter. A replacement MOSFET with a lower current rating proves to be a failure as the device fails on startup. A higher rated MOSFET resolves the problem and device begins to operate correctly.
The block diagram of the system as well as a detailed block diagram of the RF section is presented and correlated with the teardown of the RF section. The super-heterodyne down-conversion circuits and components are examined.