In this episode Shahriar takes a close look at an HP/Agilent 5347A 20GHz Frequency Counter and Power Meter. This defective unit does not provide any frequency information from Input 2 which is rated to operate up to 20GHz. Before the teardown and repair attempts, the principle operation of the instrument is reviewed.
The properties of a Step Recovery Diode (SRD) is presented along with the theory and practical aspects of generating a frequency comb. The heterodyne architecture of the frequency counter is explained in detail with the mechanism of detection and calculation of the input frequency.
During the teardown of the unit the synthesizer board, motherboard, power meter reference board and the main RF assembly are shown. The schematic of the synthesizer board and the RF board are also described. The defective component is identified and examined under the microscope. The slides for this episode can be found here.